Hi Jamie
Regarding the options to an I36NL chamber which will operate at around 25°C and 75% RH.
Based on our conversation, here are the two proposed solutions and the final pieces of information we need from you/David to complete the quote.
1. HEPA Filtration Solution
To filter the air entering the chamber when the door is opened, we recommend installing a HEPA Filtration Box in front of the unit cooler.
This setup requires the following components:
•	Specialized fans capable of tolerating a high pressure drop (similar to the Organo Genesis setup).
•	A washable, mesh-type aluminum pre-filter to catch humidifier condensate.
•	An aluminum frame filter holder.
•	A HEPA filter.
Required Action for Quote: Before I can price the HEPA filter, please confirm David’s expectation for air quality/HEPA filter specification (e.g., ISO class, minimum efficiency rating, etc.).
2. High-Temperature Decontamination
Regarding the decontamination option, we can offer a cycle that reaches 70°C by shutting down the condensing unit. Achieving anything higher than 70°C would require significant system changes and would substantially increase the cost.
Required Action for Quote: Can you/David please confirm if a 70°C decontamination temperature is sufficient for this application?
Once you confirm the required HEPA specification and the 70°C temperature acceptability, we will finalize the pricing and send the formal quote right away.
Let me know if you have any other questions!
Thank you,
Henry

